Mute design of three frequency ultrasonic cleaner
Quiet design, L series from the process and circuit architecture are designed around the quiet, we achieve more than 25 decibels lower noise than similar instruments; There is no heating. Due to the high efficiency of sound energy conversion of L series circuit and the few resonance points in the structure design, the invalid thermal energy of the whole machine is very low, which meets the requirements of temperature accuracy in the working condition. AUTO one-key automatic button design, the operation is very simple; At the same time to provide personalized selection Settings to meet the needs of various applications; LCD display interface, clear display, user operation at a glance; Chinese and English versions are optional. Two temperature displays of Fahrenheit and Celsius are optional; The ultrasonic power is adjustable in three levels; Ultrasonic single frequency, dual frequency, degassing mode is optional; Ultrasonic working time 0-60 minutes or normally open mode setting; The heating temperature range is set in the range of 0-65℃, automatically detect the body temperature; The circuit automatic frequency sweep function can greatly reduce the data error of the analyzed object or improve the efficiency of the cleaning object and the cleaning effect is more thorough.Suggest you have a look granboultrasonic cleaner. It can be applied to Metal Parts
Characteristics of three frequency ultrasonic cleaning machine
Three frequency ultrasonic cleaning method is more than the general cleaning method, especially the surface of the workpiece is more complex like some surface uneven, there are blind holes of the mechanical parts, some especially small and have higher requirements for cleanliness of products such as: watches and precision machinery parts, electronic components, circuit board components, the use of ultrasonic cleaning can achieve very ideal results.
Common problems of three frequency ultrasonic transducer
The ultrasonic oscillator is damp, you can use megohm meter to check the plug connected with the transducer, check the insulation resistance value can judge the basic situation, the general requirements of insulation resistance is greater than 5 megohm. If the insulation resistance value is not reached, the transducer is generally affected by moisture. You can put the transducer as a whole (excluding the spray shell) into the oven to set about 100℃ for drying for 3 hours or use a hair dryer to remove moisture until the resistance value is normal. When the transducer oscillator is lit and the ceramic material is broken, it can be checked with the naked eye and megohm meter. Generally, as an emergency treatment measure, the individual damaged oscillator can be disconnected, which will not affect the normal use of other oscillators. Vibrator degumming, our transducer is using cementation, screw fastening double guarantee process, in general circumstances will not appear this situation.